φ247mm x 450mm
220-240 VAC; 50 Hz
Capacity of the distilled water tank
Approx 2.5 L (Water at level Max)
Approx 0.5 L (Water at level Min)
Max. 80%, non condensing
Max. Noise level
76 kPa - 106 kPa
General Safety Instructions
-Read and understand this manual before attempting to install or operate the sterilizer.
-Make sure that all the installation conditions are fully complied with.
-Ensure that the supply voltage agrees with the supply voltage specified on the type plate of the sterilizer.
-This appliance must be grounded. Connect only to a properly grounded outlet.
-Do not cover or block any openings on this appliance.
-Use this appliance only for its intended use a described in this manual.
-Do not exceed the maximum weight limit of the loads specified in this manual.
-Do not operate this appliance if it has a damaged cord or plug, if it is not working properly, or if it has been damaged or dropped.
-Never must put into the sterilizer inflammables or explosives products.
-The sterilizer may not be operated in areas in which gas or any other explosive volatile substance is present.
-Installation and repair work should only be performed by authorized service technician. Work by unqualified persons could be dangerous and void warranty.
Position the device on a plane surface with minimum capacity 60kgs.
The sterilizer should be placed on a level worktable.
Improper water level in the chamber could cause a sterilizer malfunction.
Leave at least 10cm between the device rear part and the wall. The clearance required to open the door is 40cm.
Position the autoclave at such a height as to make it possible for the operator to check the whole sterilization chamber and carry out the normal cleaning operations.
The room where the device is installed must be enough ventilated.
Do not install the device near washing basins, taps, etc. where it is likely to be splashed.
Do not lean on the door when it is opened.
Do not place trays , papers, fluid containers, etc. on the sterilizer.